|Advance J2E Oscillator
Range: 15Hz - 50KHz in 3 switched ranges
Power: 110/240V AC 50Hz
Output: 0 - 40V P-P @5 ohm or c.50V P-P@ 600 ohm. Also has switchable attenuator
Return To Test Gear
|Hmmm... This is a nice
example of a very well built a low frequency signal
generator by Advance Instruments, England. This set goes
from dog-annoying 50KHz all the way down to gut-rumbling
15Hz - actually, I couldn't hear anything below 20Hz or
above 15KHz and had to trust a scope to tell me it was
The stability of this instrument is truly excellent, probably due to it's solid construction. An 8 hour 'soak' test at 1KHz resulted in less than 1Hz drift over the whole period. I picked this unit up very cheaply as it was described as 'non-working/no output'. I suppose most modern service techs wouldn't know a tube if it kicked 'em in the ass so I wasn't surprised to see an EL84 output pentode with a frosted getter - no vacuum. One cheap tube and 10 minutes elbow grease later - a perfectly working new toy!
The oscillator covers up to 50KHz in 3 bands, 15Hz to 300Hz, 300Hz to 4KHz & 4KHz to 50KHz. Each band has it's own scale on a large dial so it is easily set to an accurate frequency. This is enhanced by a 2 speed frequency control; turning the knob normally turns the dial at a ratio of about 5:1, but reversing it gives 1 revolution of approx 25:1 gearing before the 5:1 gear picks up again. So moving to just past the desired frequency and then reversing gives a very fine adjustment, it is quite easy to set within 1Hz at 10KHz.
There are two outputs, 5 ohm unbalanced and 600 ohm balanced, and the max output at the 600 ohm terminals is about 50v pp. The output level can be read off directly on the meter and a 20 dB attenuator can be switched into the 600 ohm output.
All in all, this instrument is considerably easier to use than the earlier model SG-66 LF Oscillator.
Return To Test Gear
Last updated 10th October 2002